Dali TE-W300H Therma Camera Setup
Firstly to setup the thermal camera, we have to setup the tripod lock on the Dali TE-W300H Thermal Camera.
Do Not This
Once done, we will have to lock the thermal camera to the tripod stand or if you don’t want to, you can skip to the next step.
Push the Free Lock outwards.
then hold the free lock and put the thermal camera on the tripod.
Now push the free lock back in to lock the Therma Camera.
To power up the Dali TE-W300H Thermal Camera, we will have to press and hold this button until a Red light pop up near the power button.
Change the color chart
To change the color chart of the Dali TE-W300H Thermal Camera ,
press the A button 1 times and press the right button in the middle to change the color chart, gradient Yellow is a recommended as it is easy reference color.
Change the measure setting
Once Done changing the Color Chart, we need to change the measure setting of the Dali TE-W300H thermal Camera, go the setting by pressing the middle button, then find the Measure, once in the Measure setting go to the body temperature correction and press the right button in the middle to enable, once done press the Middle button to confirm.
Changing the Alarm Setting
To Change the Alarm Setting, go to Menu by pressing the middle button and go to Graphic, once in the graphic setting you can scroll down and click alarm setting. In the Alarm Setting change the Alarm color to Red, now go to the Alarm Temperature and change the degrees to 37.5°C. In order to capture a picture or video if the alarm starts going off , go to AlarmLink and change to capture or record.
Capture = take a picture
Record = take a video
BlackBody is used to calibrate with Dali TE-W300H Thermal Camera.
Black Body Location
For Dali TE-W300H thermal camera to locate the BlackBody, the blackbody needs to be in the thermal camera field of vision.
To start calibrating the Dali TE-W300H Thermal Camera, press the Up button to bring up a menu.
Then choose the first option and press the middle button.
After pressing the middle button we will have to align the red cursor to the blackbody for calibration, Once the Red cursor is align u have to press the middle button, the menu will pop out again, press the C button on the thermal camera. we will now have a blue cursor pointing to the black body.
When encountered a error, do a factory reset
If encountered a error and want to do a factory reset of the Dali TE-W300H Thermal Camera, press the middle button and go to system.
When in the system setting go to Recover Default and press the middle button to do a factory reset.
Power off the Thermal Camera
To power off the Dali TE-W300H Camera. Press and hold the power button until the screen turns black
Where do i purchase Dali TE-W300H Thermal Camera
you can purchase Dali TE-W300H Thermal Camera Here